How to Verify Accuracy Using the Phantom
Hold the probe upright in the middle of the Ultrasound Registration Phantom.
Verify that the green crosshairs match the position of the white regions.
Calibration is accurate if each crosshair is directly over the white area (wires).
Check accuracy by slowly moving the probe to see if the movement of the crosshairs on the screen corresponds with the movement of the wires.
If accuracy is sufficient, select Yes to confirm.
You can now continue with your procedure.
If accuracy is insufficient, select No.
A dialog opens that provides tips for improving accuracy. Select Try again to return to the Verification page.