How to Automatically Change the Hockey Stick Probe Angle
You can verify and change the hockey stick probe angle by holding the probe near the patient reference array.
Bend the hockey stick probe angle to the required angle.
Hold the probe close to the surface of the patient reference array.
A pop-up appears, instructing the user to hold the probe tip parallel to the array surface.
Hold the probe parallel and steady.
A progress bar appears to indicate the amount of time the probe needs to be held steady.
The new set angle is confirmed visually and acoustically.