How to Perform Surface Match Registration with Z-touch
Switch on the visible laser beam following the instructions above.
Hold Z-touch perpendicular to the scanned surface so the laser beam meets the skin as a point.
Elliptical points may lead to incorrect calculation.
Using the animation as a guideline, slowly move the Z-touch laser over the green area of interest as indicated in the software.
After 60 points (half the progress bar is filled), Calculate ② is activated.
Select Start Over ① to remove all acquired points and restart registration.
Select Calculate ② to proceed immediately or acquire all points (120 points maximum) and wait for the software to proceed automatically.