The name of the product can be found on the device label.
Find your Brainlab product name and version by performing the following steps:
1. Click or press the Brainlab logo in the software (bottom right). A dialog box opens.
2. Make a note of the software name and version number from the dialog box. You need to consider the first two digits of the version to find the correct user guide.
3. Enter these details into the search field on this site. The latest version is displayed by default. If you have an earlier software version, please check the drop-down menu to see if it is available.
How to Measure Dose at a Point
Select Point under the Measure menu.
To measure the dose at a single point within an image, select a target position within the image. A point is displayed with information about the image:
Point Dose: The dose calculated at a specific point, regardless of the precalculated dose grid.
3D Volume Dose: The dose at a specific point that is calculated using a precalculated dose grid and interpolation from the surrounding points in the grid. Due to this interpolation, the actual dose value at this point might be slightly higher than the Point Dose. Therefore, the 3D Volume Dose is faster to calculate, but not as accurate as the Point Dose. The dose wash display is also interpolated from the precalculated dose grid.
To move a point, hover over it with the mouse pointer and drag it to its new position with the left mouse button.
To remove a single point, right-click on the point. To remove all points, select Point under Measure to deselect the function.
Copyright 2019, Brainlab AG Germany. All rights reserved.
Date of publication: 2019-05-22
The online version is based on the printed version of the user guide but may contain additional content.