Verify Registration Accuracy – Spine Minimally Invasive Procedure
If you are performing a minimally invasive procedure that restricts the verification of at least three landmarks, perform the following accuracy check:
Insert the instrument into the incision so that it is securely touching an anatomical landmark, reference array clamp or tooth.
Acquire fluoro images of the instrument on the bone:
Verify that the instrument position displayed on the screen corresponds to the position in the fluoro image.
After verification, two options are available. Select: