How to Set Up the Camera and Monitor
The system setup varies depending on the system used and the type of procedure being performed.
Position the camera and monitor while keeping the following points in mind:
- Select a correct camera position that will work throughout the entire procedure (including registration and navigation).
- The camera and monitor must not restrict the surgeon during surgery.
- Avoid major movement of the camera during a procedure. If the camera is moved, you must verify the accuracy again.
- The camera must have a clear view of the reference array and instruments throughout the procedure, including all registration and navigation procedures. To provide an optimum view, the camera should be approximately 1.2 – 1.8 meters (4 – 6 feet) away from the surgical field.
Plug in and turn on the system. Refer to the platform’s System User Guide (e.g., Kick, Curve).
Start the software.
Ensure the reference array is visible to the camera.
The following view is displayed to indicate whether the array is visible to the camera.
If the array is not visible, adjust the camera distance and angle until the array representation ① is visible.
Copyright 2018, Brainlab AG Germany. All rights reserved.
Publicēšanas datums: 2018-09-05