The name of the product can be found on the device label.
Find your Brainlab product name and version by performing the following steps:
1. Click or press the Brainlab logo in the software (bottom right). A dialog box opens.
2. Make a note of the software name and version number from the dialog box. You need to consider the first two digits of the version to find the correct user guide.
3. Enter these details into the search field on this site. The latest version is displayed by default. If you have an earlier software version, please check the drop-down menu to see if it is available.
Refer to the Brainlab scan protocol.
This software reads and loads approved and valid Hounsfield Unit to Electron Density (HU to ED) conversion tables only. The HU to ED conversion tables used can only be created and approved in Physics Administration.
If the scanner HU scale extends 12 bit (-1024 to +3071), the larger values are truncated:
Values < -1024 are truncated to -1024
Values > 3071 are truncated to 3071
The CT image scaling must be verified before patient treatment as only non-localized data is supported.
Import and Export Restrictions
Only data sets with a maximum scan length of 1000 mm and less than 400 slices can be imported and exported.
If objects overlap, the object with the highest HU value determines the HU value of the intersecting volume.
Copyright 2019, Brainlab AG Germany. All rights reserved.
Date of publication: 2019-05-22
The online version is based on the printed version of the user guide but may contain additional content.