DICOM Data (Third-Party Systems)
In order to import data from third-party systems into the RT Elements, the data must be pushed using a DICOM network connection using BL_IMPORT as AET at Port 104.
Any DICOM RT data that is pushed to this AET is evaluated for import conversion. For details, refer to the DICOM Conformance Statement at: http://www.brainlab.com/en/DICOM.
DICOM RT Import Performer Restrictions
The following restriction apply to imported data:
- The DICOM RT Import Performer does not import any dose or plan information that can be used by the RT Planning Elements.
- Electron-density overrides are currently not imported
- RT Structure Set contours of type POINT are currently not imported.