Distortion Correction Cranial
Certain imaging modalities are susceptible to geometric distortions arising from (e.g., system imperfections and gradient nonlinearities of the imaging system). Consequently, inaccuracies within rigid fusion results may exist after automatic fusion, manual adjustment or applying ROI fusion.
You can choose Distortion Correction Cranial to obtain a better match between images.
Image data for Distortion Correction Cranial should follow these recommendations to ensure the best results:
Image pairs should cover an intersecting volume of the patient.
A minimum of 10 slices
A slice distance lower than 4 mm (slice thickness lower than 4 mm and an acquisition without gaps are recommended)
Full DICOM information (i.e., complete DICOM header, indicating e.g., acquisition parameters)
Good raw image quality (e.g., high resolution, high contrast, minimal artifacts)
Supported Image Modalities
The following imaging modalities are supported for Distortion Correction Cranial, if paired as follows:
Unsupported Image Modalities
The following special modalities and sequence types are not supported for direct calculation:
Previously deformed image sets
FA and ADC maps
Phase and velocity maps
Gradient calibration scans
FLAWS scans (fluid and white matter suppression)
Subtraction images and projections (Minimum/Maximum Intensity Projections)
Image sets containing burned-in objects
The following content is supported and corrected based on the deformation of an image set:
Fiber bundles (e.g., DTI fiber tracts)